ViewIQ - Radiology Display Calibration Software
Display Calibration Software for Consistent Optimal Image Appearance
When diagnostic accuracy matters
Optimises the chance of detecting subtle pathologies by calibrating your image displays to radiology standards.
ViewIQ is an easy to use Microsoft Windows program for calibrating radiology and medical image displays to DICOM/ACR/RANZCR/RCR standards.
For the IT/PACS manager -Take control of your radiology monitor calibration
A web interface to your account lets you automatically collect, review and manage the calibration of a distributed fleet of radiology displays.
Central monitoring for the efficient management of your distributed fleet of displays
In a Nutshell
To maintain consistent optimal diagnostic accuracy across radiology, medical images need to be displayed consistently and optimally across radiology.
This can be achieved by the application of professional radiology standards. These standards define the display grayscale characteristics and the specifics for room lighting, display black and white levels and pixel size and screen dimensions. When applied across all equipment in the fleet from acquisition monitors to radiologists displays, consistent optimal appearance can be achieved.
X-Rite 1iDisplay PRO (1D3 OEM)
i1Display Pro Measurement Optics
|Number of Channels||3 filter design|
|Measurement area (in contact)||42mm diameter circle|
|Acceptance angle||+/- 5 degress (at half max)|
|Luminance Measurement Range||0.1 cd/m2 to 2,000 cd/m2|
|Minimum measurment integration time||0.1 sec/measurement @ 100 cd/m2|
|Maximum measurement integration time||6 sec/measurment @ 0.1 cd/m2|
|Effective measurment Primaries||CIE Standard (1931) Observer|
|Ambient measurment angle||160 degress cosine response|
|Ambient illuminance measurment range||0.5 lm/m2 to 5000 lm/m2|
i1Display Pro Measurement Performance
|Colour Accuracy||+/-0.004 x, y @ 100 cd/m2|
|Luminance Accuracy||+/- 4.0% @ 100 cd/m2|
|Color Short Term repeatability||+/- 0.001 x, y @ 100 cd/m2|
|Luminance Short-Term repeatability||+/- 1.0% @ 100 cd/m2|
- DICOM Part 14: Grayscale Standard Display Function
- ACR–AAPM–SIIM Technical Standard For Electronic Practice Of Medical Imaging
- RANZCR Standards of Practice for Clinical Radiology V11
- AAPM REPORT NO. 270 Display Quality Assurance The Report of AAPM Task Group 270 January 2019
- RCR Picture archiving and communication systems (PACS) and guidelines on diagnostic display devices, third edition
Example Professional Displays Capable of being calibrated with ViewIQ to Radiology Standards
32" General Radiology Reporting
27" General Radiology Reporting
Frequently Asked Question
ViewIQ is designed to work for the X-Rite i1Display Pro OEM version only. A firmware key prevents the software form working with the consumer version.
ViewIQ will work with Windows 10, and Windows 7. It will not work on Mac OS, iOS, or other non Windows operating systems.