ViewIQ - Radiology & Medical Display Calibration Software
Display Calibration Software for Consistent Optimal Image Appearance
Take control of your radiology & medical image monitor calibration quickly and easily.
Discover professional radiology display calibration software that’s simple to use, accurate and comprehensive all in the same package. Whether setting up a home reporting workstation or a comprehensive radiology service, we have you covered.
Solution for Your Organisation
To enquire about establishing an organization-wide solution click here to connect with our sales team.
Calibrate your radiology display in under 3 minutes
A Very Simple Test
The phrase A Very Simple Test is written with letters of reducing contrast over backgrounds increasing from black to white. If you can’t read each copy of the phrase equally well, then consider what you might also be missing from your clinical images. Each segment should look like an equal step from black through to white with no unevenly dark or light steps. If this isn’t the case or you cant see each phrase equally well ViewIQ will improve this.
*Dim room lighting to radiology reading room level for true test.
In a Nutshell
To maintain consistent optimal diagnostic accuracy across radiology, medical images need to be displayed consistently and optimally across radiology.
This can be achieved by the application of professional radiology standards. These standards define the display grayscale characteristics and the specifics for room lighting, display black and white levels and pixel size and screen dimensions. When applied across all equipment in the fleet from acquisition monitors to radiologists displays, consistent optimal appearance can be achieved.
Download ViewIQ Radiology Display Calibration Software
X-Rite 1iDisplay PRO (1D3 OEM)
i1Display Pro Measurement Optics
|Number of Channels||3 filter design|
|Measurement area (in contact)||42mm diameter circle|
|Acceptance angle||+/- 5 degress (at half max)|
|Luminance Measurement Range||0.1 cd/m2 to 2,000 cd/m2|
|Minimum measurment integration time||0.1 sec/measurement @ 100 cd/m2|
|Maximum measurement integration time||6 sec/measurment @ 0.1 cd/m2|
|Effective measurment Primaries||CIE Standard (1931) Observer|
|Ambient measurment angle||160 degress cosine response|
|Ambient illuminance measurment range||0.5 lm/m2 to 5000 lm/m2|
i1Display Pro Measurement Performance
|Colour Accuracy||+/-0.004 x, y @ 100 cd/m2|
|Luminance Accuracy||+/- 4.0% @ 100 cd/m2|
|Color Short Term repeatability||+/- 0.001 x, y @ 100 cd/m2|
|Luminance Short-Term repeatability||+/- 1.0% @ 100 cd/m2|
- DICOM Part 14: Grayscale Standard Display Function
- ACR–AAPM–SIIM Technical Standard For Electronic Practice Of Medical Imaging
- RANZCR Standards of Practice for Clinical Radiology V11
- AAPM REPORT NO. 270 Display Quality Assurance The Report of AAPM Task Group 270 January 2019
- RCR Picture archiving and communication systems (PACS) and guidelines on diagnostic display devices, third edition
Example Professional Displays Capable of being calibrated with ViewIQ to Radiology Standards
32" General Radiology Reporting
27" General Radiology Reporting
Frequently Asked Question
ViewIQ is designed to work for the X-Rite i1Display Pro OEM version only. A firmware key prevents the software form working with the consumer version.
ViewIQ will work with Windows 10, and Windows 7. It will not work on Mac OS, iOS, or other non Windows operating systems.